Maintenance / Disinfection
- Cover EndoUltra® with disposable protective barrier.
- Autoclave EndoUltra® Activator tips and handpiece sleeve at 135°C / 0.22 Mpa for 30 minutes.
- Clean handpiece with a simple disinfectant wipe.
- Monitor battery level indicator:
Frequently Asked Questions:
HOW WILL I KNOW WHEN TO CHANGE MY TIPS?
EndoUltra® Activator Tips are designed for repeated use, we estimate and recommend 20 uses per tip. At a minimum of 80 procedures per month replacing the tips at the end of each work week or every 20 procedures would be suggested. The Activator Tips are bendable to facilitate greater access. With continual bending the tip may wear at the bend joint, this will be an easily visible wear on the tip and should be replaced when this is visualized.
HOW LONG DOES THE BATTERY LAST?
The battery used in the device is a rechargeable Lithium Ion battery. The first charge is the most important and requires a 4-hour burn in. The device, like a cell phone, can be plugged in whenever not in use following the initial charge. At full charge the battery will last just over an hour of continuous run time. Considering the active run time is 60 seconds per canal this is more than enough time for a couple of days to a week worth of procedures.